This post contains some background info related to the video I posted on YouTube on how to make a simple RF current transformer, a great tool for debugging EMC / EMI issues such as radiated emissions from cables, or tracing conducted RF immunity noise paths.
RF current transformers (or probes) are commercially available products from places like Fischer CC or Solar Electronics and they work really well, have specified bandwidth and power handling characteristics, built in shielding, robust case, etc.
They also cost a few hundred £$€ each which, if you are on a budget like most people, represents a significant investment for a individual or small laboratory. However, this one can be built very cheaply; most labs will have a development kit with some clip on ferrite cores, if not the core I used only costs £5 from RS.
DIY Current Probe
I’m a big fan of making my own test adaptors and equipment as its a great way to really understand how things work and the compromises in any design. As such I decided to share how I go about making this kind of really useful tool.
It’s primary use is for A-B comparison work; measuring the current, performing a modification and then measuring the current to see the improvement.
It is to be stressed that my version is a crude but effective piece of equipment and does not replace a well designed commercial product. There’s a time and a place to invest in quality equipment and one should use engineering judgement on when that is. For instance, measuring the RF current accurately is definitely a job for a properly designed and characterised device.
Some of the design compromises involved in this low cost approach include:
Core Losses / Insertion Loss
The ferrite material in these cores is specifically designed to be lossy at the frequencies of interest, which will result in a lower reading than a higher bandwidth core and a reduction in the amount of noise on the cable downstream from the noise source. This can in some cases mask the effect you are trying to measure. The commercially available products use low loss, high bandwidth ferrite cores.
A high insertion loss also makes these parts more unsuitable for injecting noise into circuits for immunity testing. they can be calibrated for this task using a simple test setup (to be covered later)
Number of secondary turns controls sensitivity but the more you add, the inter-winding capacitance increases, decreasing the bandwidth of the tool. I generally use 5 or 6 turns to start with but I do have a 20 turn part made with micro coax on a solid core which also helps to deal with…
From the cable under test to the secondary winding. Normally a split shield (so that it doesn’t appear as a shorted turn) is built in to commercial products. Guess what, that’s easy to do on this with a spot of copper tape or foil.
Not as Robust
Although a well designed product, the plastic hinges and clips on the cores are not designed for repeated opening and closing. The Wurth Elektronik system of a special key to open and close the core is much more robust at the expense of having to keep a few keys to hand for when they inevitably go missing. However these parts are so cheap and quick to make that a broken clip on core is no real obstacle.
I’ll be following this video with some hints and tips on how to use these devices effectively for finding radiated emissions problems and for looking at conducted RF immunity issues. Stay tuned.
Video and Construction Errata
The sharp eyed of you will have spotted that I originally assembled the BNC connector on the core so that it covered the key-way to open the clamp. I rectified this but didn’t film the change.
Also, you can wrap the wire round the core without removing it from the housing but that means you don’t have a nice flat surface to affix the BNC connector to. It does make it easier to close the clamp however so make your choice.