When the customer supplied cable isn’t long enough to fit inside the standard EFT/B capacitive clamp what do you do?
One answer, for pre-compliance testing, is to make your own clamp from aluminium foil cut to length and separated from the GRP by expanded foam blocks.
The capacitive clamp is not a sophisticated piece of test equipment and a close compromise can be achieved quickly with commonly available lab materials.
Making sure there is good contact to the GRP from the generator is important which is partly achieved by taping the cable down with some conductive adhesive aluminium tape.
Overall area of the injection plate is reduced by 25% from the standard capacitive clamp plate area. Therefore the injection voltage was increased by 25% to compensate for the reduced capacitance.
Safety warning: don’t touch the foil when the generator is running!
Obviously not good enough for exact testing to the standard but it is within the spirit of the test and will give some useful information.